[SystemParameter]
RAMLength=1024	      ;RAM空间长度(k)
VoltageBase=5000
ADBits=16
ADBase=32768            ;AD分辨率

SaveType = 1 ;保存类型 short - 0 float - 1 int - 2（新软件新添加的参数）	
ContinueMaxFreq = 256000
MaxFreq = 256000
ExistExtraTrig=1
AllowFreqMode = 3
AutoCheck = 1

DetectAmplifier=0     ;是否自动检测放大器 0-设置 1-检测

SameFrequency = 0	;连续与瞬态公用频率信息
OutClkCoordXCoef = 1	;外部时钟横坐标修正系数
OutClkPntsPerCycle = 2880 ;外部时钟模式下的周期点数

GPS=0
CAN=0
ExistIO=0
PulseCounter=0
PulseCounterCount=2

ExtraSpeed0 = 0 ;控制卡附加转速模式

ExtraSpeedCount = 2	;控制卡存在几个转速通道
SplitSpeedCount = 1	;一个转速通道分几路数据，不大于8

ExtraSignal0 = 0 ;控制卡信号源模式

ExtraSignalCount = 2 ;控制卡存在几个信号源通道
SplitSignalCount = 1 ;一个信号源通道分几路数据

OutDebugInfo = 0;       输出调试信息
OutControlEnable = 1 ;  外部网络连接控制


[TriggerParameter]
TriggerLevelCount=10
TriggerLevel0=10%
TriggerLevel1=20%
TriggerLevel2=30%
TriggerLevel3=40%
TriggerLevel4=50%
TriggerLevel5=60%
TriggerLevel6=70%
TriggerLevel7=80%
TriggerLevel8=90%
TriggerLevel9=OFF

[ContinuousSampleFrequency]
FrequencyLevelCount=17
DefaultFreqIndex = 9
Frequency0  =     1Hz,  1.28Hz
Frequency1  =     2Hz,  2.56Hz
Frequency2  =     5Hz,  5.12Hz
Frequency3  =    10Hz,  12.8Hz
Frequency4  =    20Hz,  25.6Hz
Frequency5  =    50Hz,  51.2Hz
Frequency6  =   100Hz,   128Hz
Frequency7  =   200Hz,   256Hz
Frequency8  =   500Hz,   512Hz
Frequency9  =  1000Hz,  1280Hz
Frequency10 =  2000Hz,  2560Hz
Frequency11 =  5000Hz,  5120Hz
Frequency12 = 10000Hz, 12800Hz
Frequency13 = 20000Hz, 25600Hz
Frequency14 = 50000Hz, 51200Hz
Frequency15 =100000Hz,128000Hz
Frequency16 =200000Hz,256000Hz

ParaCode0  =  6,105
ParaCode1  =  6,105
ParaCode2  =  6,105
ParaCode3  =  6,105
ParaCode4  =  6,105
ParaCode5  =  5,104
ParaCode6  =  6,105
ParaCode7  =  7,106
ParaCode8  =  8,107
ParaCode9  =  9,108
ParaCode10 = 10,109
ParaCode11 = 11,110
ParaCode12 = 12,111
ParaCode13 = 13,112
ParaCode14 = 14,113
ParaCode15 = 15,114
ParaCode16 = 16,116

CtrlCode0 =5, 3
CtrlCode1 =5, 3
CtrlCode2 =5, 3
CtrlCode3 =5, 3
CtrlCode4 =5, 3
CtrlCode5 =5, 3
CtrlCode6 =5, 3
CtrlCode7 =5, 3
CtrlCode8 =4, 3
CtrlCode9 =4, 3
CtrlCode10=4, 3
CtrlCode11=4, 3
CtrlCode12=4, 3
CtrlCode13=4, 3
CtrlCode14=4, 3
CtrlCode15=4, 3
CtrlCode16=4, 3

DataRatio0 =1
DataRatio1 =1
DataRatio2 =1
DataRatio3 =1
DataRatio4 =1
DataRatio5 =1
DataRatio6 =1
DataRatio7 =1
DataRatio8 =2
DataRatio9 =4
DataRatio10=16
DataRatio11=32
DataRatio12=64
DataRatio13=200
DataRatio14=512
DataRatio15=512
DataRatio16=512

SectionNumber=7
SectionChannel0=128
SectionChannel1=64
SectionChannel2=32
SectionChannel3=16
SectionChannel4=8
SectionChannel5=4
SectionChannel6=1
SectionFrequency0=12
SectionFrequency1=13
SectionFrequency2=14
SectionFrequency3=15
SectionFrequency4=16
SectionFrequency5=16
SectionFrequency6=16

[InstantSampleFrequency]
FrequencyLevelCount = 19
DefaultFreqIndex = 9
Frequency0  =     1Hz,  1.28Hz
Frequency1  =     2Hz,  2.56Hz
Frequency2  =     5Hz,  5.12Hz
Frequency3  =     10Hz,  12.8Hz
Frequency4  =	  20Hz,  25.6Hz
Frequency5 =      50Hz,   51.2Hz
Frequency6 =     100Hz,    128Hz
Frequency7 =     200Hz,    256Hz
Frequency8 =     500Hz,    512Hz
Frequency9 =    1000Hz,   1280Hz
Frequency10 =    2000Hz,   2560Hz
Frequency11 =    5000Hz,   5120Hz
Frequency12 =   10000Hz,  12800Hz
Frequency13 =   20000Hz,  25600Hz
Frequency14 =   50000Hz,  51200Hz
Frequency15=  100000Hz, 128000Hz
Frequency16=  200000Hz, 256000Hz
Frequency17=  500000Hz, 512000Hz
Frequency18= 1000000Hz, 512000Hz

ParaCode0  =  6,105
ParaCode1  =  6,105
ParaCode2  =  6,105
ParaCode3  =  6,105
ParaCode4  =  6,105
ParaCode5 =  5,104
ParaCode6 =  6,105
ParaCode7 =  7,106
ParaCode8 =  8,107
ParaCode9 =  9,108
ParaCode10 = 10,109
ParaCode11 = 11,110
ParaCode12 = 12,111
ParaCode13 = 13,112
ParaCode14 = 14,113
ParaCode15= 15,114
ParaCode16= 16,116
ParaCode17= 17,117
ParaCode18= 18,117

CtrlCode0 =4
CtrlCode1 =4
CtrlCode2 =4
CtrlCode3 =4
CtrlCode4 =4
CtrlCode5 =4
CtrlCode6 =4
CtrlCode7 =4
CtrlCode8 =4
CtrlCode9 =4
CtrlCode10=4
CtrlCode11=4
CtrlCode12=4
CtrlCode13=4
CtrlCode14=4
CtrlCode15=4
CtrlCode16=4
CtrlCode17=4
CtrlCode18=4

[AcquisitionChannelType1]
ChannelName=DH5920
AcqTypeID = 001
FilterPosSetEna=1
FullValueCount=10
FullValue0 =20000mV, ParaCode=014
FullValue1 =10000mV, ParaCode=001
FullValue2 = 5000mV, ParaCode=002
FullValue3 = 2000mV, ParaCode=003
FullValue4 = 1000mV, ParaCode=004
FullValue5 =  500mV, ParaCode=005
FullValue6 =  200mV, ParaCode=006
FullValue7 =  100mV, ParaCode=007
FullValue8 =   50mV, ParaCode=008
FullValue9 =   20mV, ParaCode=009
FullValue10 =  10mV, ParaCode=010
FullValue11=    5mV, ParaCode=011
FullValue12=    2mV, ParaCode=012
FullValue13=    1mV, ParaCode=013

FullValueReplace7=100mV,ParaCode=020

FilterUpCount=8
FilterUpDefault = 7
FilterUp0= 10Hz, ParaCode=008
FilterUp1= 30Hz, ParaCode=007
FilterUp2=100Hz, ParaCode=006
FilterUp3=300Hz, ParaCode=005
FilterUp4= 1kHz, ParaCode=004
FilterUp5= 3kHz, ParaCode=003
FilterUp6=10kHz, ParaCode=002
FilterUp7= PASS, ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1= ON,ParaCode=001


[AcquisitionChannelType2]
ChannelName=DH5920
AcqTypeID = 002
FilterPosSetEna=1
FullValueCount=10
FullValue0 =20000mV, ParaCode=014
FullValue1 =10000mV, ParaCode=001
FullValue2 = 5000mV, ParaCode=002
FullValue3 = 2000mV, ParaCode=003
FullValue4 = 1000mV, ParaCode=004
FullValue5 =  500mV, ParaCode=005
FullValue6 =  200mV, ParaCode=006
FullValue7 =  100mV, ParaCode=020
FullValue8 =   50mV, ParaCode=008
FullValue9 =   20mV, ParaCode=009
FullValue10 =  10mV, ParaCode=010
FullValue11=    5mV, ParaCode=011
FullValue12=    2mV, ParaCode=012
FullValue13=    1mV, ParaCode=013

FilterUpCount=8
FilterUpDefault = 7
FilterUp0= 10Hz, ParaCode=008
FilterUp1= 30Hz, ParaCode=007
FilterUp2=100Hz, ParaCode=006
FilterUp3=300Hz, ParaCode=005
FilterUp4= 1kHz, ParaCode=004
FilterUp5= 3kHz, ParaCode=003
FilterUp6=10kHz, ParaCode=002
FilterUp7= PASS, ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1= ON,ParaCode=001

[AcquisitionChannelType3]
ChannelName=DH5920_tachometer
AcqTypeID = 009
FullValueCount=0
FilterUpCount=0

[AcquisitionChannelType4]
ChannelName=DH5920_Generator
AcqTypeID = 010
FullValueCount=0
FilterUpCount=0

[AcquisitionChannelType5]
ChannelName=DH5920_Counter
AcqTypeID = 050
FullValueCount=0
FilterUpCount=0

[AcquisitionChannelType6]
ChannelName=DH5939D
FilterPosSetEna=1
AcqTypeID=053
FullValueCount=10
FullValue0=20000mV,ParaCode=014
FullValue1=10000mV,ParaCode=001
FullValue2=5000mV,ParaCode=002
FullValue3=2000mV,ParaCode=003
FullValue4=1000mV,ParaCode=004
FullValue5=500mV,ParaCode=005
FullValue6=200mV,ParaCode=006
FullValue7=100mV,ParaCode=007
FullValue8=50mV,ParaCode=008
FullValue9=20mV,ParaCode=009
FullValue10=10mV,ParaCode=010
FullValue11=5mV,ParaCode=011
FullValue12=2mV,ParaCode=012
FullValue13=1mV,ParaCode=013
FullValueReplace7=100mV,ParaCode=020

FilterUpCount=8
FilterUpDefault=7
FilterUp0=100Hz, ParaCode=006
FilterUp1=300Hz, ParaCode=005
FilterUp2= 1kHz, ParaCode=004
FilterUp3= 3kHz, ParaCode=003
FilterUp4=10kHz, ParaCode=002
FilterUp5=30kHz, ParaCode=012
FilterUp6=100kHz,ParaCode=013
FilterUp7=PASS, ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1=ON,ParaCode=001

[AcquisitionChannelType7]
ChannelName=DH5920_huadongdianli_20111117
AcqTypeID = 071
FullValueCount=10
FullValue0 =30000mV, ParaCode=014
FullValue1 =10000mV, ParaCode=001
FullValue2 = 5000mV, ParaCode=002
FullValue3 = 2000mV, ParaCode=003
FullValue4 = 1000mV, ParaCode=004
FullValue5 =  500mV, ParaCode=005
FullValue6 =  200mV, ParaCode=006
FullValue7 =  100mV, ParaCode=007
FullValue8 =   50mV, ParaCode=008
FullValue9 =   20mV, ParaCode=009
FullValue10 =  10mV, ParaCode=010
FullValue11=    5mV, ParaCode=011
FullValue12=    2mV, ParaCode=012
FullValue13=    1mV, ParaCode=013

FullValueReplace7=100mV,ParaCode=020

FilterUpCount=8
FilterUpDefault = 7
FilterUp0= 10Hz, ParaCode=008
FilterUp1= 30Hz, ParaCode=007
FilterUp2=100Hz, ParaCode=006
FilterUp3=300Hz, ParaCode=005
FilterUp4= 1kHz, ParaCode=004
FilterUp5= 3kHz, ParaCode=003
FilterUp6=10kHz, ParaCode=002
FilterUp7= PASS, ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1= ON,ParaCode=001

[AcquisitionChannelType8]
ChannelName=DH5920N_shanghaijiaoda_20110627
AcqTypeID = 088
FullValueCount=10
FullValue0 =20000mV, ParaCode=014
FullValue1 =10000mV, ParaCode=001
FullValue2 = 5000mV, ParaCode=002
FullValue3 = 2000mV, ParaCode=003
FullValue4 = 1000mV, ParaCode=004
FullValue5 =  500mV, ParaCode=005
FullValue6 =  200mV, ParaCode=006
FullValue7 =  100mV, ParaCode=007
FullValue8 =   50mV, ParaCode=008
FullValue9 =   20mV, ParaCode=009
FullValue10 =  10mV, ParaCode=010
FullValue11=    5mV, ParaCode=011
FullValue12=    2mV, ParaCode=012
FullValue13=    1mV, ParaCode=013

FullValueReplace7=100mV,ParaCode=020

FilterUpCount=8
FilterUpDefault = 7
FilterUp0= 10Hz, ParaCode=008
FilterUp1=100Hz, ParaCode=006
FilterUp2=300Hz, ParaCode=005
FilterUp3= 1kHz, ParaCode=004
FilterUp4= 3kHz, ParaCode=003
FilterUp5=10kHz, ParaCode=002
FilterUp6=100kHz,ParaCode=013
FilterUp7= PASS, ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1= ON,ParaCode=001

[AcquisitionChannelType9]
ChannelName=DH5958
AcqTypeID=059
FullValueCount=10
FullValue0=20000mV,ParaCode=014
FullValue1=10000mV,ParaCode=001
FullValue2=5000mV,ParaCode=002
FullValue3=2000mV,ParaCode=003
FullValue4=1000mV,ParaCode=004
FullValue5=500mV,ParaCode=005
FullValue6=200mV,ParaCode=006
FullValue7=100mV,ParaCode=007
FullValue8=50mV,ParaCode=008
FullValue9=20mV,ParaCode=009
FullValue10=10mV,ParaCode=010
FullValue11=5mV,ParaCode=011
FullValue12=2mV,ParaCode=012
FullValue13=1mV,ParaCode=013
FullValueReplace7=100mV,ParaCode=020

FilterUpCount=8
FilterUpDefault=7
FilterUp0=10Hz,ParaCode=008
FilterUp1=30Hz,ParaCode=007
FilterUp2=100Hz,ParaCode=006
FilterUp3=300Hz,ParaCode=005
FilterUp4=1kHz,ParaCode=004
FilterUp5=3kHz,ParaCode=003
FilterUp6=10kHz,ParaCode=002
FilterUp7=PASS,ParaCode=001

AntiFilterCount=2
AntiFilter0=OFF,ParaCode=000
AntiFilter1=ON,ParaCode=001

[AmplifierChannelType1]
ChannelName=3810
AmpTypeID = 002
AmplifierCount=1
Amplifier0=1, ParaCode = 001

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004


FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=0,  ParaCode=000
Integral1=1,ParaCode=001
Integral2=2,ParaCode=002

[AmplifierChannelType2]
ChannelName=5855
AmpTypeID = 003
AmplifierCount=2
Amplifier0 = 0.1mV/pC   ,ParaCode = 001
Amplifier1 = 10mV/pC    ,ParaCode = 002


FilterUpCount=0
FilterLowCount=0

IntegralCount=0
CapacitorType = 1Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType3]
ChannelName=5856_1Hz
AmpTypeID = 004
AmplifierCount=2
Amplifier0 =  1  ,ParaCode = 001
Amplifier1 = 50  ,ParaCode = 006
;AmplifierCount=3
;Amplifier0 =  1  ,ParaCode = 001
;Amplifier1 = 10  ,ParaCode = 004
;Amplifier2 =100  ,ParaCode = 007

FilterUpCount=0
IntegralCount=3
CapacitorType = 1Hz
Integral0=0,ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType4]
ChannelName=5855-1
AmpTypeID = 005
AmplifierCount=2
Amplifier0 = 0.1mV/pC   ,ParaCode = 001
Amplifier1 = 10mV/pC    ,ParaCode = 002

FilterUpCount=0
FilterLowCount=0

IntegralCount=3
CapacitorType = 1Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003


[AmplifierChannelType5]
ChannelName=3810-200
AmpTypeID = 002
AmplifierCount=1
Amplifier0=200, ParaCode = 010

BridgeVoltageCount=1 ;恒压
BridgeVoltage0= 2V, ParaCode=001


[AmplifierChannelType6]
ChannelName=3810-500
AmpTypeID=072
AmplifierCount=1
Amplifier0=500,ParaCode=012

BridgeVoltageCount=4 
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=10V,ParaCode=003
BridgeVoltage3=24V,ParaCode=004


[AmplifierChannelType7]
ChannelName=3812  ;双恒流适调器
AmpTypeID = 007
AmplifierCount=1
Amplifier0=1, ParaCode = 001

BridgeMode = 1  ;供桥模式 0 - 横压  1 - 恒流
ConstantCurrentCount = 4 ;恒流
ConstantCurrent0 = 1mA,ParaCode=0
ConstantCurrent1 = 2mA,ParaCode=0
ConstantCurrent2 = 5mA,ParaCode=0
ConstantCurrent3 = 10mA,ParaCode=0

[AmplifierChannelType8]
ChannelName=3810  ;应变50倍增益适调器
AmpTypeID = 008
AmplifierCount=1
Amplifier0=50, ParaCode = 008

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType9]
ChannelName=3811  ;电流测量适调器
AmpTypeID = 006
AmplifierCount=1
Amplifier0=1, ParaCode = 001

[AmplifierChannelType10]
ChannelName=5856_10Hz
AmpTypeID = 014
;AmplifierCount=2
;Amplifier0 =  1  ,ParaCode = 001
;Amplifier1 = 50  ,ParaCode = 006
AmplifierCount=3
Amplifier0 =  1  ,ParaCode = 001
Amplifier1 = 10  ,ParaCode = 004
Amplifier2 =100  ,ParaCode = 007

FilterUpCount=0
IntegralCount=3
CapacitorType = 10Hz
Integral0=  0,ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType11]
ChannelName=3810  ;应变200倍增益适调器
AmpTypeID = 015
AmplifierCount=1
Amplifier0=200, ParaCode = 010

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType12]
ChannelName=3810_100  ;应变100倍增益适调器
AmpTypeID = 022
AmplifierCount=1
Amplifier0=100, ParaCode = 009

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType13]
ChannelName=3810_2000  ;应变2000倍增益适调器
AmpTypeID = 023
AmplifierCount=1
Amplifier0=2000, ParaCode = 014

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType14]
ChannelName=3814_Pt100  
AmpTypeID = 013
AmplifierCount=1
Amplifier0=1, ParaCode = 001

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType15]
ChannelName=5856-3_0.1Hz
AmpTypeID = 046
AmplifierCount=1
Amplifier0 =  1  ,ParaCode = 001

FilterUpCount=0
IntegralCount=2
CapacitorType = 0.1Hz
Integral0=  0,ParaCode=001
Integral1=1,ParaCode=002

[AmplifierChannelType16]
ChannelName=3840s
AmpTypeID=054
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=30,ParaCode=007

BridgeVoltageCount=8 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2= 6V, ParaCode=006
BridgeVoltage3= 8V, ParaCode=007
BridgeVoltage4=10V, ParaCode=003
BridgeVoltage5=12V, ParaCode=008
BridgeVoltage6=16V, ParaCode=009
BridgeVoltage7=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=0,  ParaCode=000
Integral1=1,ParaCode=001
Integral2=2,ParaCode=002

[AmplifierChannelType17]
ChannelName=3810_100  ;应变100倍增益适调器
AmpTypeID = 055
AmplifierCount=1
Amplifier0=100, ParaCode = 009

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType18]
ChannelName=5857
AmpTypeID = 063
AmplifierCount=2
Amplifier0 =0.01mV/pC ,ParaCode = 003
Amplifier1 =  10mV/pC ,ParaCode = 002


FilterUpCount=0
FilterLowCount=0

IntegralCount=0
CapacitorType = 1Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType19]
ChannelName=3814_B;624suo 
AmpTypeID = 066
AmplifierCount=1
Amplifier0=100, ParaCode = 009

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType20]
ChannelName=3840s_63839_504
AmpTypeID=067
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=100,ParaCode=009

BridgeVoltageCount=4;恒压
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=10V,ParaCode=003
BridgeVoltage3=24V,ParaCode=004

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=0,ParaCode=000
Integral1=1,ParaCode=001
Integral2=2,ParaCode=002

[AmplifierChannelType21]
ChannelName=3840s_701_20101115
AmpTypeID=070
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=50,ParaCode=008

BridgeVoltageCount=4;恒压
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=10V,ParaCode=003
BridgeVoltage3=24V,ParaCode=004

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=0,ParaCode=000
Integral1=1,ParaCode=001
Integral2=2,ParaCode=002

[AmplifierChannelType22]
ChannelName=5855-1
AmpTypeID = 074
AmplifierCount=2
Amplifier0 = 0.0025mV/pC,ParaCode = 009
Amplifier1 = 10mV/pC    ,ParaCode = 002

FilterUpCount=0
FilterLowCount=0

IntegralCount=3
CapacitorType = 1Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType23]
ChannelName=5855-1
AmpTypeID = 075
AmplifierCount=2
Amplifier0 = 0.1mV/pC   ,ParaCode = 001
Amplifier1 = 10mV/pC    ,ParaCode = 002


FilterUpCount=0
FilterLowCount=0

IntegralCount=3
CapacitorType = 10Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType24]
ChannelName=3810N_100  ;应变100倍增益适调器
AmpTypeID = 086
AmplifierCount=1
Amplifier0=100, ParaCode = 009

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

BridgeSetMode=1
CalibrationMode=1

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType25]
ChannelName=5856_1Hz
AmpTypeID=089
AmplifierCount=3
Amplifier0=1,ParaCode=001
Amplifier1=10,ParaCode=004
Amplifier2=100,ParaCode=007

FilterUpCount=0
IntegralCount=3
CapacitorType=1Hz
Integral0=无积分,ParaCode=001
Integral1=一次积分,ParaCode=002
Integral2=二次积分,ParaCode=003

[AmplifierChannelType26]
ChannelName=3810E
AmpTypeID=091
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=100,ParaCode=009

BridgeVoltageCount=8
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=6V,ParaCode=006
BridgeVoltage3=10V,ParaCode=003
BridgeVoltage4=12V,ParaCode=008
BridgeVoltage5=15V,ParaCode=005
BridgeVoltage6=20V,ParaCode=011
BridgeVoltage7=24V,ParaCode=004

BridgeSetMode=1
CalibrationMode=1

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType27]
ChannelName=5857-2
AmpTypeID = 03
AmpSubVersion=03
AmplifierCount=2
Amplifier0 =0.001mV/pC ,ParaCode = 003
Amplifier1 =  1mV/pC ,ParaCode = 002

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
CapacitorType = 1Hz 	;
Integral0=0,  ParaCode=001
Integral1=1,ParaCode=002
Integral2=2,ParaCode=003

[AmplifierChannelType28]
ChannelName=3810A
AmpTypeID=052
AmplifierCount=1
Amplifier0=0.4,ParaCode=000

BridgeVoltageCount=0
BridgeVoltage0=24V,ParaCode=004

[AmplifierChannelType29]
ChannelName=3840s_701
AmpTypeID=061
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=100,ParaCode=009

BridgeVoltageCount=8
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2= 6V, ParaCode=006
BridgeVoltage3= 8V, ParaCode=007
BridgeVoltage4=10V, ParaCode=003
BridgeVoltage5=12V, ParaCode=008
BridgeVoltage6=16V, ParaCode=009
BridgeVoltage7=24V, ParaCode=004

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=无积分,  ParaCode=000
Integral1=一次积分,ParaCode=001
Integral2=二次积分,ParaCode=002

[AmplifierChannelType30]
ChannelName=3840s_507;2012-03-14
AmpTypeID=099
AmplifierCount=2
Amplifier0=1,ParaCode=001
Amplifier1=100,ParaCode=007

BridgeVoltageCount=8
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=6V,ParaCode=006
BridgeVoltage3=8V,ParaCode=007
BridgeVoltage4=10V,ParaCode=003
BridgeVoltage5=12V,ParaCode=008
BridgeVoltage6=16V,ParaCode=009
BridgeVoltage7=24V,ParaCode=004

FilterUpCount=0
FilterLowCount=0

IntegralCount=0
Integral0=无积分,ParaCode=000
Integral1=一次积分,ParaCode=001
Integral2=二次积分,ParaCode=002

[AmplifierChannelType31]
ChannelName=3810T
AmpTypeID=106
AmplifierCount=11
Amplifier0=1,ParaCode=001
Amplifier1=1,ParaCode=002
Amplifier2=1,ParaCode=004
Amplifier3=1,ParaCode=005
Amplifier4=1,ParaCode=006
Amplifier5=1,ParaCode=008
Amplifier6=100,ParaCode=009
Amplifier7=100,ParaCode=010
Amplifier8=100,ParaCode=012
Amplifier9=100,ParaCode=013
Amplifier10=100,ParaCode=014

BridgeVoltageCount=8
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=6V,ParaCode=006
BridgeVoltage3=10V,ParaCode=003
BridgeVoltage4=12V,ParaCode=008
BridgeVoltage5=15V,ParaCode=005
BridgeVoltage6=20V,ParaCode=011
BridgeVoltage7=24V,ParaCode=004

BridgeSetMode=1
CalibrationMode=1

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType32]
ChannelName=3810N_100
AmpTypeID=125
AmplifierCount=1
Amplifier0=100,ParaCode=009

BridgeVoltageCount=4
BridgeVoltage0=2V,ParaCode=001
BridgeVoltage1=5V,ParaCode=002
BridgeVoltage2=10V,ParaCode=003
BridgeVoltage3=24V,ParaCode=004
BridgeSetMode=1
CalibrationMode=1
FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType33]
ChannelName=3810N_2  ;
AmpTypeID = 011
AmplifierCount=1
Amplifier0=1, ParaCode = 001

BridgeVoltageCount=4 ;恒压
BridgeVoltage0= 2V, ParaCode=001
BridgeVoltage1= 5V, ParaCode=002
BridgeVoltage2=10V, ParaCode=003
BridgeVoltage3=24V, ParaCode=004

BridgeSetMode=1
CalibrationMode=1

FilterUpCount=0
FilterLowCount=0

[AmplifierChannelType34]
ChannelName=5857_1
AmpTypeID = 063
AmpSubVersion=17
AmplifierCount=2
Amplifier0 =0.01mV/pC ,ParaCode = 003

[InputModeItem]
InputModeCount=5
InputMode0=DIF_DC,ParaCode=001
InputMode1=SIN_DC,ParaCode=002
InputMode2=    AC,ParaCode=003
InputMode3=  IEPE,ParaCode=004
InputMode4=   GND,ParaCode=005

[FilterModeItem]
MeasureContent14=1
MeasureContent23=1
MeasureContent24=1
MeasureContent25=1
InputModeContent4=1

[ChannelType0100]
ChannelName=DH5920
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 2  ; 测量内容项
MeasureSel0 =  0
MeasureSel1 =  6


InputModeCount=5
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode2 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode3 = 3,AcqInputModeIndex = 3,AmpInputModeIndex = -1
InputMode4 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1


FullValueCount=10
DefaultFullValue=10000
FullValue0 =20000mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue2 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue3 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue4 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =  200mV, AcqIndex= 6, AmpIndex = 0
FullValue7 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue8 =   50mV, AcqIndex= 8, AmpIndex = 0
FullValue9 =   20mV, AcqIndex= 9, AmpIndex = 0
;FullValue10 =  10mV, AcqIndex=10, AmpIndex = 0
;FullValue11=    5mV, AcqIndex=11, AmpIndex = 0
;FullValue12=    2mV, AcqIndex=12, AmpIndex = 0
;FullValue13=    1mV, AcqIndex=13, AmpIndex = 0

[ChannelType0300]
ChannelName=DH5920_tachometer
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 = 7
MeasureSel1 = 18
InputModeCount=0
FullValueCount=0

[ChannelType0400]
ChannelName=DH5920_Generator
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 =  8
InputModeCount=0
FullValueCount=0

[ChannelType0500]
ChannelName=DH5920_Generator
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 = 17
InputModeCount=0
FullValueCount=0

[ChannelType0101]
ChannelName=DH5920_3810
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=3  ; 测量内容项
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue1 =   10mV, AcqIndex=10, AmpIndex = 0
FullValue2 =    1mV, AcqIndex=13, AmpIndex = 0



[ChannelType0102]
ChannelName=DH5920_5855
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4  =     10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0103]
ChannelName=DH5920_5856_1Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

;FullValueCount=10
;FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
;FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
;FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
;FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
;FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
;FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
;FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
;FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
;FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
;FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2

FullValueCount=10   
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue4 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue5 =  200mV, AcqIndex= 1, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 2, AmpIndex = 1
FullValue7 =   40mV, AcqIndex= 3, AmpIndex = 1
FullValue8 =   20mV, AcqIndex= 4, AmpIndex = 1
FullValue9 =   10mV, AcqIndex= 5, AmpIndex = 1


[ChannelType0104]
ChannelName=DH5920_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4 =      10pC,  AcqIndex = 7, AmpIndex = 1


[ChannelType0105]
ChannelName=DH5920_3810-200
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=8
FullValue0 = 100mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =  50mV, AcqIndex= 1, AmpIndex = 0
FullValue2 =  20mV, AcqIndex= 2, AmpIndex = 0
FullValue3 =  10mV, AcqIndex= 3, AmpIndex = 0
FullValue4 =   5mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =   2mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =   1mV, AcqIndex= 6, AmpIndex = 0
FullValue7 = 0.5mV, AcqIndex= 7, AmpIndex = 0


[ChannelType0106]
ChannelName=DH5920_3810-500
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0=20mV, AcqIndex= 1, AmpIndex = 0
FullValue1=2mV, AcqIndex= 4, AmpIndex = 0
FullValue2=0.2mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2=1

[ChannelType0107]
ChannelName=DH5920_3812
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=3  ; 测量内容项
MeasureSel0=5
MeasureSel1=2
MeasureSel2=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=7
FullValue0 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue1 =   50mV, AcqIndex= 8, AmpIndex = 0
FullValue2 =   20mV, AcqIndex= 9, AmpIndex = 0
FullValue3 =   10mV, AcqIndex=10, AmpIndex = 0
FullValue4 =    5mV, AcqIndex=11, AmpIndex = 0
FullValue5 =    2mV, AcqIndex=12, AmpIndex = 0
FullValue6 =    1mV, AcqIndex=13, AmpIndex = 0

[ChannelType0108]
ChannelName=DH5920_3810_50
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 = 100mV, AcqIndex= 2, AmpIndex = 0
FullValue1 =  10mV, AcqIndex= 5, AmpIndex = 0
FullValue2 =   1mV, AcqIndex= 8, AmpIndex = 0

[ChannelType0109]
ChannelName=DH5920_3811
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 =  15

InputModeCount=2
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=1
FullValue0 = 5000mV, AcqIndex= 2, AmpIndex = 0

[ChannelType0110]
ChannelName=DH5920_5856_10Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2

[ChannelType0111]
ChannelName=DH5920_3810_200
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=4
FullValue0 =  100mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =   10mV, AcqIndex= 3, AmpIndex = 0
FullValue2 =    1mV, AcqIndex= 6, AmpIndex = 0
FullValue3 =  0.1mV, AcqIndex= 9, AmpIndex = 0

[ChannelType0112]
ChannelName=DH5920_3810_100
VltNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0113]
ChannelName=DH5920_3810_2000
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =  1mV, AcqIndex= 3, AmpIndex = 0
FullValue1 =0.5mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =0.1mV, AcqIndex= 6, AmpIndex = 0

[ChannelType0114]
ChannelName=DH5923_3814
BalanceOption = 0    ;允许进行通道平衡标志
AcqType=0
AmpType=-1
MeasureSelCount = 1  ; 测量内容项(默认选择的通道)
MeasureSel0 = 5

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=5
FullValue0 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue1 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue2 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue3 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue4 =  200mV, AcqIndex= 6, AmpIndex = 0

[ChannelType0115]
ChannelName=DH5920_5856-3_0.1Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=2
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue4 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue5 =  200mV, AcqIndex= 6, AmpIndex = 0
FullValue6 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue7 =   50mV, AcqIndex= 8, AmpIndex = 0
FullValue8 =   20mV, AcqIndex= 9, AmpIndex = 0
FullValue9 =   10mV, AcqIndex= 10, AmpIndex =0

[ChannelType0117]
ChannelName=DH5920_3810_100
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=4
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode2 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0118]
ChannelName=DH5920_5857
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0  =1000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  = 100000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =  10000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue5  =     10pC,  AcqIndex = 7, AmpIndex = 1


[ChannelType0119]
ChannelName=DH5920_3814B
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 2  ; 测量内容项
MeasureSel0 =  0
MeasureSel1 =  6

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=4
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 50mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 20mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 10mV, AcqIndex= 4, AmpIndex = 0

[ChannelType0120]
ChannelName=DH5920N_3840s_63839_504
BalanceOption=1;BalanceOption
AcqType=0
AmpType=-1
MeasureSelCount=3;测量内容项
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6
MeasureSel3=18

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=0,AmpInputModeIndex=4

FullValueCount=13
FullValue0=10000mV,AcqIndex=1,AmpIndex=0
FullValue1=5000mV,AcqIndex=2,AmpIndex=0
FullValue2=2000mV,AcqIndex=3,AmpIndex=0
FullValue3=1000mV,AcqIndex=4,AmpIndex=0
FullValue4=500mV,AcqIndex=5,AmpIndex=0
FullValue5=200mV,AcqIndex=6,AmpIndex=0
FullValue6=100mV,AcqIndex=1,AmpIndex=1
FullValue7=50mV,AcqIndex=2,AmpIndex=1
FullValue8=20mV,AcqIndex=3,AmpIndex=1
FullValue9=10mV,AcqIndex=4,AmpIndex=1
FullValue10=5mV,AcqIndex=5,AmpIndex=1
FullValue11=2mV,AcqIndex=6,AmpIndex=1
FullValue12=1mV,AcqIndex=7,AmpIndex=1
FullValueReplace12=1

[ChannelType0121]
ChannelName=DH5920N_3840s_701
BalanceOption=1;BalanceOption
AcqType=0
AmpType=-1
MeasureSelCount=3;测量内容项
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6
MeasureSel3=18

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=0,AmpInputModeIndex=4

FullValueCount=13
FullValue0=10000mV,AcqIndex=1,AmpIndex=0
FullValue1=5000mV,AcqIndex=2,AmpIndex=0
FullValue2=2000mV,AcqIndex=3,AmpIndex=0
FullValue3=1000mV,AcqIndex=4,AmpIndex=0
FullValue4=500mV,AcqIndex=5,AmpIndex=0
FullValue5=200mV,AcqIndex=1,AmpIndex=1
FullValue6=100mV,AcqIndex=2,AmpIndex=1
FullValue7=40mV,AcqIndex=3,AmpIndex=1
FullValue8=20mV,AcqIndex=4,AmpIndex=1
FullValue9=10mV,AcqIndex=5,AmpIndex=1
FullValue10=4mV,AcqIndex=6,AmpIndex=1
FullValue11=2mV,AcqIndex=7,AmpIndex=1
FullValue12=1mV,AcqIndex=8,AmpIndex=1
FullValueReplace11=1

[ChannelType0122]
ChannelName=DH5920_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0 = 4000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1 =  400000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2 =   40000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3 =    1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue5 =      10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0123]
ChannelName=DH5923_5855_10Hz
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4 =      10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0124]
ChannelName=DH5920_3810N_100
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0125]
ChannelName=DH5920_5856_1Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2

[ChannelType0126]
ChannelName=DH5920_3810E_100
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = 0
InputMode1 = 4,AcqInputModeIndex = 0,AmpInputModeIndex = 4
ShortMode = 4,AcqInputModeIndex = 0,AmpInputModeIndex = 4

FullValueCount=14
DefaultFullValue=10000
FullValue0 =20000mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue2 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue3 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue4 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =  200mV, AcqIndex= 6, AmpIndex = 0
FullValue7 =  100mV, AcqIndex= 1, AmpIndex = 1
FullValue8 =   50mV, AcqIndex= 2, AmpIndex = 1
FullValue9 =   20mV, AcqIndex= 3, AmpIndex = 1
FullValue10 =  10mV, AcqIndex=4, AmpIndex = 1
FullValue11=    5mV, AcqIndex=5, AmpIndex = 1
FullValue12=    2mV, AcqIndex=6, AmpIndex = 1
FullValue13=    1mV, AcqIndex=7, AmpIndex = 1
FullValueReplace13 = 1

[ChannelType0127]
ChannelName=DH5920_5857_2
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0  =10000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  = 1000000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =  100000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3  =   10000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4  =    1000pC,  AcqIndex = 4, AmpIndex = 1
FullValue5  =     100pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0133]
ChannelName=DH5920_3810N_2
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=5
FullValue0 =2000mV, AcqIndex= 3, AmpIndex = 0
FullValue1 =1000mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =500mV, AcqIndex= 5, AmpIndex = 0
FullValue3 =200mV, AcqIndex= 6, AmpIndex = 0
FullValue4 =100mV, AcqIndex= 7, AmpIndex = 0

[ChannelType0134]
ChannelName=DH5920_5857
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 2
FullValue0  =1000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  = 100000pC,  AcqIndex = 4, AmpIndex = 0

[ChannelType0600]
ChannelName=DH5920N
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=0
MeasureSel1=6
MeasureSel2=5

InputModeCount=5
InputMode0=0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1=1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode2=2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode3=3,AcqInputModeIndex = 3,AmpInputModeIndex = -1
InputMode4=4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode=4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=10
DefaultFullValue=10000
FullValue0=20000mV, AcqIndex=0, AmpIndex=0
FullValue1=10000mV, AcqIndex=1, AmpIndex=0
FullValue2=5000mV, AcqIndex=2, AmpIndex=0
FullValue3=2000mV, AcqIndex=3, AmpIndex=0
FullValue4=1000mV, AcqIndex=4, AmpIndex=0
FullValue5=500mV, AcqIndex=5, AmpIndex=0
FullValue6=200mV, AcqIndex=6, AmpIndex=0
FullValue7=100mV, AcqIndex=7, AmpIndex=0
FullValue8=50mV, AcqIndex=8, AmpIndex=0
FullValue9=20mV, AcqIndex=9, AmpIndex=0

[ChannelType0601]
ChannelName=DH5920N_3810
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=3  
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=3
FullValue0=100mV,AcqIndex= 7, AmpIndex = 0
FullValue1=10mV,AcqIndex=10, AmpIndex = 0
FullValue2=1mV,AcqIndex=13, AmpIndex = 0



[ChannelType0602]
ChannelName=DH5920N_5855
AcqType = 0
AmpType = 1
MeasureSelCount = 1
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4  =     10pC,  AcqIndex = 7, AmpIndex = 1


[ChannelType0603]
ChannelName=DH5920N_5856_1Hz
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=1  
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1


FullValueCount=10   
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue4 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue5 =  200mV, AcqIndex= 1, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 2, AmpIndex = 1
FullValue7 =   50mV, AcqIndex= 3, AmpIndex = 1
FullValue8 =   20mV, AcqIndex= 4, AmpIndex = 1
FullValue9 =   10mV, AcqIndex= 5, AmpIndex = 1


[ChannelType0604]
ChannelName=DH5920N_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5

FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4 =      10pC,  AcqIndex = 7, AmpIndex = 1


[ChannelType0605]
ChannelName=DH5920N_3810-200
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=8
FullValue0 = 100mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =  50mV, AcqIndex= 1, AmpIndex = 0
FullValue2 =  20mV, AcqIndex= 2, AmpIndex = 0
FullValue3 =  10mV, AcqIndex= 3, AmpIndex = 0
FullValue4 =   5mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =   2mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =   1mV, AcqIndex= 6, AmpIndex = 0
FullValue7 = 0.5mV, AcqIndex= 7, AmpIndex = 0


[ChannelType0606]
ChannelName=DH5920N_3810-500
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=5
FullValue0=20mV, AcqIndex= 1, AmpIndex = 0
FullValue1=10mV, AcqIndex= 2, AmpIndex = 0
FullValue2=4mV, AcqIndex= 3, AmpIndex = 0
FullValue3=2mV, AcqIndex= 4, AmpIndex = 0
FullValue4=1mV, AcqIndex= 5, AmpIndex = 0

[ChannelType0607]
ChannelName=DH5920N_3812
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=3  
MeasureSel0=2
MeasureSel1=4
MeasureSel2=5

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=7
FullValue0 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue1 =   50mV, AcqIndex= 8, AmpIndex = 0
FullValue2 =   20mV, AcqIndex= 9, AmpIndex = 0
FullValue3 =   10mV, AcqIndex=10, AmpIndex = 0
FullValue4 =    5mV, AcqIndex=11, AmpIndex = 0
FullValue5 =    2mV, AcqIndex=12, AmpIndex = 0
FullValue6 =    1mV, AcqIndex=13, AmpIndex = 0

[ChannelType0608]
ChannelName=DH5920N_3810_50
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 = 100mV, AcqIndex= 2, AmpIndex = 0
FullValue1 =  10mV, AcqIndex= 5, AmpIndex = 0
FullValue2 =   1mV, AcqIndex= 8, AmpIndex = 0


[ChannelType0609]
ChannelName=DH5920_3811
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount = 1  
MeasureSel0 =  15

InputModeCount=2
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=1
FullValue0 = 5000mV, AcqIndex= 2, AmpIndex = 0

[ChannelType0610]
ChannelName=DH5920_5856_10Hz
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=1 
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2

[ChannelType0611]
ChannelName=DH5920_3810_200
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2  
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=4
FullValue0 =  100mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =   10mV, AcqIndex= 3, AmpIndex = 0
FullValue2 =    1mV, AcqIndex= 6, AmpIndex = 0
FullValue3 =  0.1mV, AcqIndex= 9, AmpIndex = 0

[ChannelType0612]
ChannelName=DH5920_3810_100
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0613]
ChannelName=DH5920N_3810_2000
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2  
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =  1mV, AcqIndex= 3, AmpIndex = 0
FullValue1 =0.5mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =0.1mV, AcqIndex= 6, AmpIndex = 0

[ChannelType0614]
ChannelName=DH5920N_3814
BalanceOption=0
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=5

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModelndex = -1
ShortMode = 4,AcqInputModelIndex = 4,AmpInputModelndex = -1

FullValueCount=2
FullValue0=500mv,AcqIndex=5,AmpIndex=0
FullValue1=200mv,AcqIndex=6,AmpIndex=0

[ChannelType0615]
ChannelName=DH5920N_5856-3_0.1Hz
BalanceOption=1
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=0

InputModeCount=2
InputMode0=1,AcqInputModeIndex=1,AmpInputModeIndex=1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=1
shortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=1

FullValueCount=10
FullValue0=10000mv,AcqIndex=1,AmpIndex=0
FullValue1=5000mv,AcqIndex=2,AmpIndex=0
FullValue2=2000mv,AcqIndex=3,AmpIndex=0
FullValue3=1000mv,AcqIndex=4,AmpIndex=0
FullValue4=500mv,AcqIndex=5,AmpIndex=0
FullValue5=200mv,AcqIndex=6,AmpIndex=0
FullValue6=100mv,AcqIndex=7,AmpIndex=0
FullValue7=50mv,AcqIndex=8,AmpIndex=0
FullValue8=20mv,AcqIndex=9,AmpIndex=0
FullValue9=10mv,AcqIndex=10,AmpIndex=0

[ChannelType0618]
ChannelName=DH592N_5857
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0  =1000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  = 100000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =  10000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue5  =     10pC,  AcqIndex = 7, AmpIndex = 1


[ChannelType0620]
ChannelName=DH5920N_3840s_63839_504
BalanceOption=1;BalanceOption
AcqType=0
AmpType=-1
MeasureSelCount=3;测量内容项
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6
MeasureSel3=18

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=0,AmpInputModeIndex=4

FullValueCount=13
FullValue0=10000mV,AcqIndex=1,AmpIndex=0
FullValue1=5000mV,AcqIndex=2,AmpIndex=0
FullValue2=2000mV,AcqIndex=3,AmpIndex=0
FullValue3=1000mV,AcqIndex=4,AmpIndex=0
FullValue4=500mV,AcqIndex=5,AmpIndex=0
FullValue5=200mV,AcqIndex=6,AmpIndex=0
FullValue6=100mV,AcqIndex=1,AmpIndex=1
FullValue7=50mV,AcqIndex=2,AmpIndex=1
FullValue8=20mV,AcqIndex=3,AmpIndex=1
FullValue9=10mV,AcqIndex=4,AmpIndex=1
FullValue10=5mV,AcqIndex=5,AmpIndex=1
FullValue11=2mV,AcqIndex=6,AmpIndex=1
FullValue12=1mV,AcqIndex=7,AmpIndex=1
FullValueReplace12=1

[ChannelType0621]
ChannelName=DH5920N_3840s_701
BalanceOption=1;BalanceOption
AcqType=0
AmpType=-1
MeasureSelCount=3;测量内容项
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6
MeasureSel3=18

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=0,AmpInputModeIndex=4

FullValueCount=13
FullValue0=10000mV,AcqIndex=1,AmpIndex=0
FullValue1=5000mV,AcqIndex=2,AmpIndex=0
FullValue2=2000mV,AcqIndex=3,AmpIndex=0
FullValue3=1000mV,AcqIndex=4,AmpIndex=0
FullValue4=500mV,AcqIndex=5,AmpIndex=0
FullValue5=200mV,AcqIndex=1,AmpIndex=1
FullValue6=100mV,AcqIndex=2,AmpIndex=1
FullValue7=40mV,AcqIndex=3,AmpIndex=1
FullValue8=20mV,AcqIndex=4,AmpIndex=1
FullValue9=10mV,AcqIndex=5,AmpIndex=1
FullValue10=4mV,AcqIndex=6,AmpIndex=1
FullValue11=2mV,AcqIndex=7,AmpIndex=1
FullValue12=1mV,AcqIndex=8,AmpIndex=1
FullValueReplace11=1

[ChannelType0624]
ChannelName=DH5920_3810_100
StrainNegativePolarity=1
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0625]
ChannelName=DH5920_5856_1Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2

[ChannelType0626]
ChannelName=DH5920_3810_100
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0700]
ChannelName=DH5920
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 =  0
MeasureSel1 =  6
MeasureSel2 =  5


InputModeCount=5
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode2 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode3 = 3,AcqInputModeIndex = 3,AmpInputModeIndex = -1
InputMode4 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1


FullValueCount=10
FullValue0 =30000mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue2 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue3 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue4 = 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =  500mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =  200mV, AcqIndex= 6, AmpIndex = 0
FullValue7 =  100mV, AcqIndex= 7, AmpIndex = 0
FullValue8 =   50mV, AcqIndex= 8, AmpIndex = 0
FullValue9 =   20mV, AcqIndex= 9, AmpIndex = 0

[ChannelType0709]
ChannelName=DH5920_3811
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 =  15

InputModeCount=2
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=1
FullValue0 = 5000mV, AcqIndex= 2, AmpIndex = 0

[ChannelType0710]
ChannelName=DH5920_5856_10Hz
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=1  ; 测量内容项
MeasureSel0=0

InputModeCount=4
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2 = 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10
FullValue0 =10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2 = 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3 = 1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4 =  500mV, AcqIndex= 2, AmpIndex = 1
FullValue5 =  200mV, AcqIndex= 3, AmpIndex = 1
FullValue6 =  100mV, AcqIndex= 1, AmpIndex = 2
FullValue7 =   50mV, AcqIndex= 2, AmpIndex = 2
FullValue8 =   20mV, AcqIndex= 3, AmpIndex = 2
FullValue9 =   10mV, AcqIndex= 4, AmpIndex = 2


[ChannelType0622]
ChannelName=DH5920N_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0 = 4000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1 =  400000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2 =   40000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3 =    1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue5 =      10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0800]
ChannelName=DH5920N
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=0
MeasureSel1=6
MeasureSel2=5

InputModeCount=5
InputMode0=0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1=1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode2=2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode3=3,AcqInputModeIndex = 3,AmpInputModeIndex = -1
InputMode4=4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode=4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=10
DefaultFullValue=10000
FullValue0=20000mV, AcqIndex=0, AmpIndex=0
FullValue1=10000mV, AcqIndex=1, AmpIndex=0
FullValue2=5000mV, AcqIndex=2, AmpIndex=0
FullValue3=2000mV, AcqIndex=3, AmpIndex=0
FullValue4=1000mV, AcqIndex=4, AmpIndex=0
FullValue5=500mV, AcqIndex=5, AmpIndex=0
FullValue6=200mV, AcqIndex=6, AmpIndex=0
FullValue7=100mV, AcqIndex=7, AmpIndex=0
FullValue8=50mV, AcqIndex=8, AmpIndex=0
FullValue9=20mV, AcqIndex=9, AmpIndex=0

[ChannelType0802]
ChannelName=DH5920_5855
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4  =     10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0804]
ChannelName=DH5920N_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1
MeasureSel0=3


InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5

FullValue0  = 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  =  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3 =     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4 =      10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0812]
ChannelName=DH5920N_3810_100
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0818]
ChannelName=DH5920_5857
AcqType = 0
AmpType = 1
MeasureSelCount = 1;测量内容项
MeasureSel0=3

InputModeCount=2
InputMode0 = 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 6
FullValue0  =1000000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1  = 100000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2  =  10000pC,  AcqIndex = 7, AmpIndex = 0
FullValue3  =   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue4  =    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue5  =     10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0900]
ChannelName=DH5958
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=2
MeasureSel0=0
MeasureSel1=14

InputModeCount=5
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=1,AcqInputModeIndex=1,AmpInputModeIndex=-1
InputMode2=2,AcqInputModeIndex=2,AmpInputModeIndex=-1
InputMode3=3,AcqInputModeIndex=3,AmpInputModeIndex=-1
InputMode4=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

MeasureType14_InputModeCount=2
MeasureType14_InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
MeasureType14_InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
MeasureType14_ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=10
DefaultFullValue=10000
FullValue0=20000mV,AcqIndex=0,AmpIndex=0
FullValue1=10000mV,AcqIndex=1,AmpIndex=0
FullValue2=5000mV,AcqIndex=2,AmpIndex=0
FullValue3=2000mV,AcqIndex=3,AmpIndex=0
FullValue4=1000mV,AcqIndex=4,AmpIndex=0
FullValue5=500mV,AcqIndex=5,AmpIndex=0
FullValue6=200mV,AcqIndex=6,AmpIndex=0
FullValue7=100mV,AcqIndex=7,AmpIndex=0
FullValue8=50mV,AcqIndex=8,AmpIndex=0
FullValue9=20mV,AcqIndex=9,AmpIndex=0

[ChannelType0901]
ChannelName=DH5920_3810
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=3  
MeasureSel0=2
MeasureSel1=4
MeasureSel2=6

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=3
FullValue0=100mV,AcqIndex= 7, AmpIndex = 0
FullValue1=10mV,AcqIndex=10, AmpIndex = 0
FullValue2=1mV,AcqIndex=13, AmpIndex = 0

[ChannelType0902]
ChannelName=DH5920_5855
AcqType = 0
AmpType = 1
MeasureSelCount = 1
MeasureSel0=3

InputModeCount=2
InputMode0= 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0= 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1=  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2=   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3=    100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4=     10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0903]
ChannelName=DH5920_5856_1Hz
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=1  
MeasureSel0=0

InputModeCount=4
InputMode0= 1,AcqInputModeIndex = 1,AmpInputModeIndex = 1
InputMode1= 2,AcqInputModeIndex = 2,AmpInputModeIndex = 1
InputMode2= 3,AcqInputModeIndex = 2,AmpInputModeIndex = 3
InputMode3= 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1
ShortMode= 4,AcqInputModeIndex = 4,AmpInputModeIndex = 1

FullValueCount=10   
FullValue0=10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1= 5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2= 2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3= 1000mV, AcqIndex= 4, AmpIndex = 0
FullValue4=  500mV, AcqIndex= 5, AmpIndex = 0
FullValue5=  200mV, AcqIndex= 1, AmpIndex = 1
FullValue6=  100mV, AcqIndex= 2, AmpIndex = 1
FullValue7=   40mV, AcqIndex= 3, AmpIndex = 1
FullValue8=   20mV, AcqIndex= 4, AmpIndex = 1
FullValue9=   10mV, AcqIndex= 5, AmpIndex = 1

[ChannelType0904]
ChannelName=DH5920_5855-1
AcqType = 0
AmpType = 1
MeasureSelCount = 1
MeasureSel0=3

InputModeCount=2
InputMode0= 2,AcqInputModeIndex = 2,AmpInputModeIndex = -1
InputMode1= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount = 5
FullValue0= 100000pC,  AcqIndex = 1, AmpIndex = 0
FullValue1=  10000pC,  AcqIndex = 4, AmpIndex = 0
FullValue2=   1000pC,  AcqIndex = 1, AmpIndex = 1
FullValue3=     100pC,  AcqIndex = 4, AmpIndex = 1
FullValue4=      10pC,  AcqIndex = 7, AmpIndex = 1

[ChannelType0905]
ChannelName=DH5920_3810-200
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=8
FullValue0 = 100mV, AcqIndex= 0, AmpIndex = 0
FullValue1 =  50mV, AcqIndex= 1, AmpIndex = 0
FullValue2 =  20mV, AcqIndex= 2, AmpIndex = 0
FullValue3 =  10mV, AcqIndex= 3, AmpIndex = 0
FullValue4 =   5mV, AcqIndex= 4, AmpIndex = 0
FullValue5 =   2mV, AcqIndex= 5, AmpIndex = 0
FullValue6 =   1mV, AcqIndex= 6, AmpIndex = 0
FullValue7 = 0.5mV, AcqIndex= 7, AmpIndex = 0

[ChannelType0906]
ChannelName=DH5920_3810-500
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0= 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0=20mV, AcqIndex= 1, AmpIndex = 0
FullValue1=2mV, AcqIndex= 4, AmpIndex = 0
FullValue2=0.2mV, AcqIndex= 7, AmpIndex = 0

[ChannelType0907]
ChannelName=DH5920_3812
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=3  
MeasureSel0=2
MeasureSel1=4
MeasureSel2=5

InputModeCount=2
InputMode0= 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode= 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=7
FullValue0=100mV, AcqIndex= 7, AmpIndex= 0
FullValue1=50mV, AcqIndex= 8, AmpIndex= 0
FullValue2=20mV, AcqIndex= 9, AmpIndex= 0
FullValue3=10mV, AcqIndex=10, AmpIndex= 0
FullValue4=5mV, AcqIndex=11, AmpIndex= 0
FullValue5=2mV, AcqIndex=12, AmpIndex= 0
FullValue6=1mV, AcqIndex=13, AmpIndex= 0

[ChannelType0908]
ChannelName=DH5920_3810_50
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0= 0,AcqInputModeIndex= 0,AmpInputModeIndex= -1
InputMode1= 4,AcqInputModeIndex= 4,AmpInputModeIndex= -1
ShortMode= 4,AcqInputModeIndex= 4,AmpInputModeIndex= -1

FullValueCount=3
FullValue0=100mV, AcqIndex= 2, AmpIndex= 0
FullValue1=10mV, AcqIndex= 5, AmpIndex= 0
FullValue2=1mV, AcqIndex= 8, AmpIndex= 0

[ChannelType0909]
ChannelName=DH5920_3811
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount = 1  ; 测量内容项
MeasureSel0 =  15

InputModeCount=2
InputMode0 = 1,AcqInputModeIndex = 1,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=1
FullValue0 = 5000mV, AcqIndex= 2, AmpIndex = 0

[ChannelType0910]
ChannelName=DH5920_5856_10Hz
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=1 
MeasureSel0=0

InputModeCount=4
InputMode0= 1,AcqInputModeIndex= 1,AmpInputModeIndex= 1
InputMode1= 2,AcqInputModeIndex= 2,AmpInputModeIndex= 1
InputMode2= 3,AcqInputModeIndex= 2,AmpInputModeIndex= 3
InputMode3= 4,AcqInputModeIndex= 4,AmpInputModeIndex= 1
ShortMode= 4,AcqInputModeIndex= 4,AmpInputModeIndex= 1

FullValueCount=10
FullValue0=10000mV, AcqIndex= 1, AmpIndex = 0
FullValue1=5000mV, AcqIndex= 2, AmpIndex = 0
FullValue2=2000mV, AcqIndex= 3, AmpIndex = 0
FullValue3=1000mV, AcqIndex= 1, AmpIndex = 1
FullValue4=500mV, AcqIndex= 2, AmpIndex = 1
FullValue5=200mV, AcqIndex= 3, AmpIndex = 1
FullValue6=100mV, AcqIndex= 1, AmpIndex = 2
FullValue7=50mV, AcqIndex= 2, AmpIndex = 2
FullValue8=20mV, AcqIndex= 3, AmpIndex = 2
FullValue9=10mV, AcqIndex= 4, AmpIndex = 2

[ChannelType0911]
ChannelName=DH5920_3810_200
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2  
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=4
FullValue0=100mV,AcqIndex=0,AmpIndex=0
FullValue1=10mV,AcqIndex=3,AmpIndex=0
FullValue2=1mV,AcqIndex=6,AmpIndex=0
FullValue3=0.1mV,AcqIndex=9,AmpIndex=0

[ChannelType0912]
ChannelName=DH5923N_3810_100
VltNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0913]
ChannelName=DH5920_3810_2000
BalanceOption = 1 
AcqType=0
AmpType=-1
MeasureSelCount=2  
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=3
FullValue0=1mV,AcqIndex=3,AmpIndex=0
FullValue1=0.5mV,AcqIndex=4,AmpIndex=0
FullValue2=0.1mV,AcqIndex=6,AmpIndex=0

[ChannelType0914]
ChannelName=DH5923_3814
BalanceOption=0
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=5

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModelndex=-1
ShortMode=4,AcqInputModelIndex=4,AmpInputModelndex=-1

FullValueCount=2
FullValue0=500mv,AcqIndex=5,AmpIndex=0
FullValue1=200mv,AcqIndex=6,AmpIndex=0

[ChannelType0915]
ChannelName=DH5920_5856-3_0.1Hz
BalanceOption=1
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=0

InputModeCount=2
InputMode0=1,AcqInputModeIndex=1,AmpInputModeIndex=1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=1
shortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=1

FullValueCount=10
FullValue0=10000mv,AcqIndex=1,AmpIndex=0
FullValue1=5000mv,AcqIndex=2,AmpIndex=0
FullValue2=2000mv,AcqIndex=3,AmpIndex=0
FullValue3=1000mv,AcqIndex=4,AmpIndex=0
FullValue4=500mv,AcqIndex=5,AmpIndex=0
FullValue5=200mv,AcqIndex=6,AmpIndex=0
FullValue6=100mv,AcqIndex=7,AmpIndex=0
FullValue7=50mv,AcqIndex=8,AmpIndex=0
FullValue8=20mv,AcqIndex=9,AmpIndex=0
FullValue9=10mv,AcqIndex=10,AmpIndex=0

[ChannelType0924]
ChannelName=DH5923N_3810N_100
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=3
FullValue0 =100mV, AcqIndex= 1, AmpIndex = 0
FullValue1 = 10mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =  1mV, AcqIndex= 7, AmpIndex = 0
FullValueReplace2 = 1

[ChannelType0926]
ChannelName=DH5958_3810E
StrainNegativePolarity=1
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=2  
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=4

FullValueCount=14
DefaultFullValue=10000
FullValue0=20000mV,AcqIndex=0,AmpIndex=0
FullValue1=10000mV,AcqIndex=1,AmpIndex=0
FullValue2=5000mV,AcqIndex=2,AmpIndex=0
FullValue3=2000mV,AcqIndex=3,AmpIndex=0
FullValue4=1000mV,AcqIndex=4,AmpIndex=0
FullValue5=500mV,AcqIndex=5,AmpIndex=0
FullValue6=200mV,AcqIndex=6,AmpIndex=0
FullValue7=100mV,AcqIndex=1,AmpIndex=1
FullValue8=50mV,AcqIndex=2,AmpIndex=1
FullValue9=20mV,AcqIndex=3,AmpIndex=1
FullValue10=10mV,AcqIndex=4,AmpIndex=1
FullValue11=5mV,AcqIndex=5,AmpIndex=1
FullValue12=2mV,AcqIndex=6,AmpIndex=1
FullValue13=1mV,AcqIndex=7,AmpIndex=1
FullValueReplace13=1

[ChannelType0928]
ChannelName=DH5920_3810A
BalanceOption=0
AcqType=0
AmpType=-1
MeasureSelCount=1
MeasureSel0=0

InputModeCount=3
InputMode0=1,AcqInputModeIndex=1,AmpInputModeIndex=-1
InputMode1=2,AcqInputModeIndex=2,AmpInputModeIndex=-1
InputMode2=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=4
FullValue0=25000mV,AcqIndex=1,AmpIndex=0
FullValue1=12500mV,AcqIndex=2,AmpIndex=0
FullValue2=5000mV,AcqIndex=3,AmpIndex=0
FullValue3=2500mV,AcqIndex=4,AmpIndex=0

[ChannelType0930]
ChannelName=DH5958_3840S_100
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=2
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=4

FullValueCount=14
DefaultFullValue=10000
FullValue0=20000mV,AcqIndex=0,AmpIndex=0
FullValue1=10000mV,AcqIndex=1,AmpIndex=0
FullValue2=5000mV,AcqIndex=2,AmpIndex=0
FullValue3=2000mV,AcqIndex=3,AmpIndex=0
FullValue4=1000mV,AcqIndex=4,AmpIndex=0
FullValue5=500mV,AcqIndex=5,AmpIndex=0
FullValue6=200mV,AcqIndex=6,AmpIndex=0
FullValue7=100mV,AcqIndex=1,AmpIndex=1
FullValue8=50mV,AcqIndex=2,AmpIndex=1
FullValue9=20mV,AcqIndex=3,AmpIndex=1
FullValue10=10mV,AcqIndex=4,AmpIndex=1
FullValue11=5mV,AcqIndex=5,AmpIndex=1
FullValue12=2mV,AcqIndex=6,AmpIndex=1
FullValue13=1mV,AcqIndex=7,AmpIndex=1
FullValueReplace13=1

[ChannelType0931]
ChannelName=DH5958_3840T_100
BalanceOption=1  
AcqType=0
AmpType=-1
MeasureSelCount=2
MeasureSel0=4
MeasureSel1=2

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=0
InputMode1=4,AcqInputModeIndex=0,AmpInputModeIndex=4
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=4

FullValueCount=11
FullValue0=10000mV,AcqIndex=1,AmpIndex=0
FullValue1=5000mV,AcqIndex=2,AmpIndex=1
FullValue2=2000mV,AcqIndex=3,AmpIndex=2
FullValue3=1000mV,AcqIndex=4,AmpIndex=3
FullValue4=500mV,AcqIndex=5,AmpIndex=4
FullValue5=200mV,AcqIndex=6,AmpIndex=5
FullValue6=100mV,AcqIndex=1,AmpIndex=6
FullValue7=50mV,AcqIndex=2,AmpIndex=7
FullValue8=20mV,AcqIndex=3,AmpIndex=8
FullValue9=10mV,AcqIndex=4,AmpIndex=9
FullValue10=5mV,AcqIndex=5,AmpIndex=10

[ChannelType0932]
ChannelName=DH5958_3810N_100
StrainNegativePolarity=1
BalanceOption = 1  
AcqType=0
AmpType=-1
MeasureSelCount=2 
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0=0,AcqInputModeIndex=0,AmpInputModeIndex=-1
InputMode1=4,AcqInputModeIndex=4,AmpInputModeIndex=-1
ShortMode=4,AcqInputModeIndex=4,AmpInputModeIndex=-1

FullValueCount=3
FullValue0=100mV,AcqIndex=1,AmpIndex=0
FullValue1=10mV,AcqIndex=4,AmpIndex=0
FullValue2=1mV,AcqIndex=7,AmpIndex=0
FullValueReplace2=1

StrainBridgeTypeCount=7
StrainBridgeType0=方式1(120Ω),BridgeModeIndex=6
StrainBridgeType1=方式1(350Ω),BridgeModeIndex=7
StrainBridgeType2=方式2,BridgeModeIndex=1
StrainBridgeType3=方式3,BridgeModeIndex=2
StrainBridgeType4=方式4,BridgeModeIndex=3
StrainBridgeType5=方式5,BridgeModeIndex=4
StrainBridgeType6=方式6,BridgeModeIndex=5

[ChannelType0933]
ChannelName=DH5920_3810N_2
StrainNegativePolarity=1
BalanceOption = 1  ;Balance Option
AcqType=0
AmpType=-1
MeasureSelCount=2  ; 测量内容项
MeasureSel0=2
MeasureSel1=4

InputModeCount=2
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1
InputMode1 = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1
ShortMode = 4,AcqInputModeIndex = 4,AmpInputModeIndex = -1

FullValueCount=5
FullValue0 =2000mV, AcqIndex= 3, AmpIndex = 0
FullValue1 =1000mV, AcqIndex= 4, AmpIndex = 0
FullValue2 =500mV, AcqIndex= 5, AmpIndex = 0
FullValue3 =200mV, AcqIndex= 6, AmpIndex = 0
FullValue4 =100mV, AcqIndex= 7, AmpIndex = 0

[Channel]
ACQChannelTypeCount=9 ;数采通道类型数
AMPChannelTypeCount=34 ;适配器通道类型数
ChannelFirst=0
ChannelCount=56
ChannelsPerCase=16
ChannelTypeCount=3
ChannelType0=0100
ChannelType1=0114
ChannelType2=0110
ChannelType3=0104
ChannelType4=0104
ChannelType5=0101
ChannelType6=0100
ChannelType7=0100
ChannelType8=0100
ChannelType9=0100
ChannelType10=0100
ChannelType11=0100
ChannelType12=0100
ChannelType13=0100
ChannelType14=0100
ChannelType15=0100
ChannelType16=0100
ChannelType17=0100
ChannelType18=0100
ChannelType19=0100
ChannelType20=0100
ChannelType21=0100
ChannelType22=0100
ChannelType23=0100
ChannelType24=0100
ChannelType25=0100
ChannelType26=0100
ChannelType27=0100
ChannelType28=0100
ChannelType29=0100
ChannelType30=0100
ChannelType31=0100
ChannelType32=0100
ChannelType33=0100
ChannelType34=0100
ChannelType35=0100
ChannelType36=0100
ChannelType37=0100
ChannelType38=0100
ChannelType39=0100
ChannelType40=0100
ChannelType41=0100
ChannelType42=0100
ChannelType43=0100
ChannelType44=0100
ChannelType45=0100
ChannelType46=0100
ChannelType47=0100
ChannelType48=0100
ChannelType49=0100
ChannelType50=0100
ChannelType51=0100
ChannelType52=0100
ChannelType53=0100
ChannelType54=0100
ChannelType55=0100
ChannelType56=0100
ChannelType57=0100
ChannelType58=0100
ChannelType59=0100
ChannelType60=0100
ChannelType61=0100
ChannelType62=0100
ChannelType63=0100
ChannelType64=0100
ChannelType65=0100
ChannelType66=0100
ChannelType67=0100
ChannelType68=0100
ChannelType69=0100
ChannelType70=0100
ChannelType71=0100
ChannelType72=0100
ChannelType73=0100
ChannelType74=0100
ChannelType75=0100
ChannelType76=0100
ChannelType77=0100
ChannelType78=0100
ChannelType79=0100
ChannelType80=0100
ChannelType81=0100
ChannelType82=0100
ChannelType83=0100
ChannelType84=0100
ChannelType85=0100
ChannelType86=0100
ChannelType87=0100
ChannelType88=0100
ChannelType89=0100
ChannelType90=0100
ChannelType91=0100
ChannelType92=0100
ChannelType93=0100
ChannelType94=0100
ChannelType95=0100
ChannelType96=0100
ChannelType97=0100
ChannelType98=0100
ChannelType99=0100
ChannelType100=0100
ChannelType101=0100
ChannelType102=0100
ChannelType103=0100
ChannelType104=0100
ChannelType105=0100
ChannelType106=0100
ChannelType107=0100
ChannelType108=0100
ChannelType109=0100
ChannelType110=0100
ChannelType111=0100
ChannelType112=0100
ChannelType113=0100
ChannelType114=0100
ChannelType115=0100
ChannelType116=0100
ChannelType117=0100
ChannelType118=0100
ChannelType119=0100
ChannelType120=0100
ChannelType121=0100
ChannelType122=0100
ChannelType123=0100
ChannelType124=0100
ChannelType125=0100
ChannelType126=0100
ChannelType127=0100
ChannelType128=0100
ChannelType129=0100
ChannelType130=0100
ChannelType131=0100
ChannelType132=0100
ChannelType133=0100
ChannelType134=0100
ChannelType135=0100
ChannelType136=0100
ChannelType137=0100
ChannelType138=0100
ChannelType139=0100
ChannelType140=0100
ChannelType141=0100
ChannelType142=0100
ChannelType143=0100
ChannelType144=0100
ChannelType145=0100
ChannelType146=0100
ChannelType147=0100
ChannelType148=0100
ChannelType149=0100
ChannelType150=0100
ChannelType151=0100
ChannelType152=0100
ChannelType153=0100
ChannelType154=0100
ChannelType155=0100
ChannelType156=0100
ChannelType157=0100
ChannelType158=0100
ChannelType159=0100
ChannelType160=0100
ChannelType161=0100
ChannelType162=0100
ChannelType163=0100
ChannelType164=0100
ChannelType165=0100
ChannelType166=0100
ChannelType167=0100
ChannelType168=0100
ChannelType169=0100
ChannelType170=0100
ChannelType171=0100
ChannelType172=0100
ChannelType173=0100
ChannelType174=0100
ChannelType175=0100
ChannelType176=0100
ChannelType177=0100
ChannelType178=0100
ChannelType179=0100
ChannelType180=0100
ChannelType181=0100
ChannelType182=0100
ChannelType183=0100
ChannelType184=0100
ChannelType185=0100
ChannelType186=0100
ChannelType187=0100
ChannelType188=0100
ChannelType189=0100
ChannelType190=0100
ChannelType191=0100
ChannelType192=0100
ChannelType193=0100
ChannelType194=0100
ChannelType195=0100
ChannelType196=0100
ChannelType197=0100
ChannelType198=0100
ChannelType199=0100
ChannelType200=0100
ChannelType201=0100
ChannelType202=0100
ChannelType203=0100
ChannelType204=0100
ChannelType205=0100
ChannelType206=0100
ChannelType207=0100
ChannelType208=0100
ChannelType209=0100
ChannelType210=0100
ChannelType211=0100
ChannelType212=0100
ChannelType213=0100
ChannelType214=0100
ChannelType215=0100
ChannelType216=0100
ChannelType217=0100
ChannelType218=0100
ChannelType219=0100
ChannelType220=0100
ChannelType221=0100
ChannelType222=0100
ChannelType223=0100
ChannelType224=0100
ChannelType225=0100
ChannelType226=0100
ChannelType227=0100
ChannelType228=0100
ChannelType229=0100
ChannelType230=0100
ChannelType231=0100
ChannelType232=0100
ChannelType233=0100
ChannelType234=0100
ChannelType235=0100
ChannelType236=0100
ChannelType237=0100
ChannelType238=0100
ChannelType239=0100
ChannelType240=0100
ChannelType241=0100
ChannelType242=0100
ChannelType243=0100
ChannelType244=0100
ChannelType245=0100
ChannelType246=0100
ChannelType247=0100
ChannelType248=0100
ChannelType249=0100
ChannelType250=0100
ChannelType251=0100
ChannelType252=0100
ChannelType253=0100
ChannelType254=0100
ChannelType255=0100
