[SystemParameter]
VoltageBase = 20000
ADBits = 24
ADBase = 8388608            ;AD分辨率
SameFrequency = 1	;连续与瞬态公用频率信息
SaveType = 1 ;保存类型 short - 0 float - 1 int - 2
ContinueMaxFreq = 5;硬件能够使用连续采集的最大频率
MaxFreq = 5; 实时回收数据的最大连续采样频率
AllowFreqMode = 1
IgnorePacket = 0; 是否忽略包号，不回收
PacketMax255 = 1; 采样过程中是否最大包号为255 老仪器默认1
[ContinuousSampleFrequency]
DefaultFreqIndex = 0
FrequencyLevelCount = 3
Frequency0 = 1Hz,  1.28Hz
Frequency1 = 2Hz,  2.56Hz
Frequency2 = 5Hz,  5.12Hz

ParaCode0 = 0,0
ParaCode1 = 1,1
ParaCode2 = 2,2

SectionNumber = 1
SectionVersion1 = 1
SectionChannel0 = 128
SectionFrequency0 = 3
SectionNumber1 = 1
SectionChannel1_0 = 128
SectionFrequency1_0 = 3

[InputModeItem]
InputModeCount=5
InputMode0=DIF_DC,ParaCode=001
InputMode1=SIN_DC,ParaCode=002
InputMode2=    AC,ParaCode=003
InputMode3=  IEPE,ParaCode=004
InputMode4=   GND,ParaCode=005

[AcquisitionChannelType1]
ChannelName = DH5905I
BridgeSetMode = 1
AcqTypeID = 200
FullValueCount = 1
FullValue0 = 23.4375mV, ParaCode=009
BridgeVoltageCount = 1
BridgeVoltage0 = 3V,ParaCode=010

[ChannelType0100]
ChannelName = DH5905I
CalibrationMode = 1
StrainNegativePolarity = 1
AcqType = 0
AmpType = -1
MeasureSelCount = 1
MeasureSel0 = 4
FullValueCount = 1
FullValue0 = 20mV,AcqIndex=0,AmpIndex=0
InputModeCount=1
InputMode0 = 0,AcqInputModeIndex = 0,AmpInputModeIndex = -1

[Channel]
ACQChannelTypeCount = 1 ;数采通道类型数
AMPChannelTypeCount = 0 ;适配器通道类型数
ChannelFirst = 0
ChannelCount = 1024
ChannelsPerCase = 1
ChannelTypeCount = 1
ChannelType0=0100
ChannelType1=0100
ChannelType2=0100
ChannelType3=0100
ChannelType4=0100
ChannelType5=0100
ChannelType6=0100
ChannelType7=0100
